The ANENG AN8008 / AN8009 multimeters have 3.6×10 mm ceramic fuses on their inputs:
Based on past experience, at some point over the next year or five, I’ll forget to plug the hot probe back in the voltage hole before measuring a power supply:
Whereupon the fuse will blow.
So, for about five bucks, a bag of 10 A and 0.5 A axial lead fast-blow glass fuses just arrived from halfway around the planet:
They have the right body size and, in this application, fine points concerning current ratings and cartridge composition don’t make much difference. If I actually need one, I’ll snip off the leads, jam it in the holder, and move on.