Archive for July 23rd, 2018
The maximum current drops from 20 A for frequencies above 20 kHz:
There’s a 100 A·μs pulse charge limit:
Because the probe is actually a pulse transformer, its internal termination imposes a (small) load on the input circuit:
The specs are 100 mΩ at 1 MHz and 500 mΩ at 50 MHz, which means the load is essentially zilch for the circuits and signals I deal with.
The Tektronix Probes for Current Measurement Systems has some useful descriptions.
Memo to Self: Should any of those limits matter, rethink what you’re doing.
An interesting story about the AM503 design from someone who lived through it.